This standard specifies accelerated test method for light resistance (*1) of non-metallic materials and products in automobile cabin using light-exposure apparatus with xenon-arc lamp (*2) (hereinafter call “Test apparatus”).
(*1) The property of resistance to variation of color, gloss, etc. under irradiation of light, and shall be distinguished from weather resistance, which is the property of resistance to variation to natural actions, such as out-door sunlight, wind, rain, etc.
(*2) Test apparatus capable of performing irradiation of an artificial light source closely resembling sunlight for examining part of the properties of light resistance in a short time.
Definitions of main terms used in this standard shall be accordance with Table 1.
1) Black panel temperature
(hereinafter call “B.P.T”)
The temperature indicated by a black panel thermometer attached to a rack or rotary board, which is representative for the surface temperature of test pieces.
2) Xenon-arc lamp
An arc lamp which emits by xenon-arc discharge.
3) Emission part
The part composed of a xenon-arc lamp and a filter.
For evaluating variation of color, gloss, etc., conditions, such as temperature, humidity, irradiance, etc., which are effective on test results, shall be in accordance with Table 2.
2) Relative humidity
4) Irradiance ratio
Irradiance of 320nm max. shall be under 1.5% of that of 300-400nm
5) Irradiation method
6) Test period control
According to radiant exposure quantity
Note(*3): Selection of irradiance shall be effective on acceleration, however, irradiance also shall be effective on sample temperature and heat history. Appropriate irradiance is needed to be selected considering effect on sample by temperature.
Test apparatus shall be constructed of an emission part, xenon-arc starting and stabilizing apparatus, irradiance adjusting apparatus, temperature adjust apparatus, test tank, humidity adjusting apparatus, test piece holder, cooling apparatus, instrumentation apparatus, etc.
4.1 Emission Part
4.1.1 Emission Part shall be as follows, according to clause 3:
1) Xenon-arc lamp: Xenon-arc lamps are long arc lamps of a.c. burning in which electrodes are sealed-in both ends of a quartz tube and xenon gas is sealed-in inside, and are used with performing forced water cooling or forced air cooling.
2) Filter: Filters are used for obtaining the required spectral characteristics and for cooling the xenon-arc lamp. For insulating the ultraviolet part, glass filters for ultraviolet-radiation insulation, running water, etc., are used.
3) Other: Based on the portion to be used, material superior in heat resistance, corrosion resistance, humidity resistance, or electric insulation shall be used.
4.1.2 Spectral characteristics of emission part
The xenon-arc lamp shall have a continuous spectrum at least over the wavelength range of 250 to 1200 nm, and shall have a similar spectral characteristics to glass-through-sunlight in combination with filter.
4.1.3 Irradiance of test piece surface
Irradiance of test piece surfaces shall be able to be set on any certain value within 48-100W/m2 at least over the wavelength range of 300 to 400 nm.
Also, spectral irradiances of test piece surfaces in the wavelength of 300 to 400nm shall be indicated for each type of test apparatus. Further, the irradiance in other wavelength ranges or the spectral irradiance at s specific wavelength should also be marked upon agreement between the parties concerned. The measuring method is to be in accordance with the annex of JIS B 7754.
4.1.4 Tolerance in irradiance during operation
It shall be capable of maintaining ±10% of the set irradiance.
4.1.5 Tolerance in irradiance of test pieces surfaces
The difference of irradiances of test piece surfaces due to the position must not exceed 10% of the maximum value.
4.1.6 Irradiation method
As a rule, a certain constant irradiance shall be kept continuously during operation.
4.2 Irradiance Adjusting Apparatus
The irradiance adjusting apparatus is an apparatus to correct the attenuation of radiation energy accompanying the use of a xenon-arc lamp and filter, measuring the irradiance in the wavelength of 300 to 400 nm, and shall be provided with one of the following conditions:
1) It shall be capable of manual adjustment of the lamp power to make the irradiance constant.
2) It shall be capable of automatic control of the lamp power to make the irradiance constant.
3) It shall be capable of manual adjustment of the lamp power stepwise accoding to the time of use.
4.3Black Panel Thermometer
Black panel thermometer is so structured that the heat sensitive body, bimetal, platinum resistor, thermistor, thermoelectric couple, etc., is attached coincident with the center of the metal plate and is adhered tightly to the heat sensitive body protecting tube.
Remarks: Structures and dimensions shall be in accordance with 5.7(2) and Attached Fig.1. Type A of JIS B 7754.
4.4 Test Tank
Temperature and humidity in test tank shall be as follows:
1) B.P.T. at the position of test piece shall be 89±3℃.
Remarks: Temperature will be different depending on the combination of xenon-arc lamp type and filter.
2) Relative humidity
It shall favorably be capable of keeping relative humidity 50±5% under the temperature specified in 1). Further, relative humidity shall be measured at the point close to and inside the exit of test tank.
Items unspecified in the foregoing clauses, such as test tank, xenon-arc starting and stabilizing apparatus, temperature adjusting apparatus, humidity adjusting apparatus, test piece holder, cooling apparatus, instrumentation apparatus, etc., shall be in accordance with JIB B 7754.
5.1Dimensions of Test Piece
Irradiation area of test piece shall be 30 mm min in vertical direction and over 30mm min horizontal direction, and thickness shall be determined according to agreement to agreement between the parties concerned. Further, when test piece has urethane foam backing layer as part construction, density and composition, etc. of the layer shall be determined between the parties concerned.
In case of test piece with texture direction, such as fabrics, etc., attachment direction relative to test apparatus shall be indicated.
6.1 Attach test piece on test piece holder firmly so that it won’t drop. Further, when backing material like urethane foam is attached on backside of test piece, the backside shall be contracted with backing material as tightly as possible so that it shall not become loose.
6.2Set test piece holders with test pieces attached on a rack or a rotary board without gaps between the holders. When there are some test piece holders without test pieces, put white-colored papers, aluminum plates, stainless steel plates, etc., on all of the holder in order to keep heat equilibrium inside test tank.
6.3 Select a certain irradiance within 48-100W/m2 in the wavelength of 300-400nm, and adjust test apparatus so that the selected irradiance shall be kept constant during exposure.
6.4 Adjust B.P.T during exposure within 89±3℃ by temperature adjusting apparatus.
6.5 Adjust humidity during exposure within 50±5% RH.
6.6 In case of broad samples, their position shall favorably be replaced at certain intervals for better test accuracy.
Remarks: Radiant exposure quantity can be calculated by the following formula:
Radiant exposure quantity (KJ/m2) = Irradiance (W/m2)×Time(Hr)×3.6
JIS B 7754
JASO M 346
以上标准信息来自互联网搜索整理，如需了解完整的HES D6601-99A的信息或HES D6601-99A标准解读，欢迎致电【400-6808-138】详询。
Q-SUN Xe-1-B是一款适合小型实验室的氙灯老化箱,其命名规程B指的是改型号的产品具有背部喷淋功能，这意味着在SAE部分标准的测试中尤其有用。Q-SUN Xe-1-B 马上询价
Q-SUN Xe-3-HSE氙灯老化试验箱具有全光谱光照功能、湿度控制以及喷淋功能，其命名规则为湿度控制(H)、喷淋(S)。Q-SUN Xe-3-HSE 马上询价
Q-sun Xe-1-BCE氙灯加速老化试验箱配有全光谱氙弧灯、太阳眼辐照度控制系统，该型号的氙灯加速老化试验箱同时具有背部喷功能。Q-sun Xe-1-BCE 马上询价
Q-SUN Xe-1-SCE氙灯耐候试验机是一款支持水喷淋功能和制冷功能的氙灯加速老化设备Q-SUN Xe-1-SCE 马上询价
Q-SUN Xe-3-HCE是一款具有全光谱光照模拟、湿度控制以及制冷功能的氙灯老化试验机,其命名规则为湿度控制（H）、制冷功能（C）。Q-SUN Xe-3-HCE 马上询价
Q-SUN Xe-3-HDSE氙灯耐气候试验箱具有湿度控制、以及双喷淋功能,其命名规则为湿度(H)、双喷淋(D)、喷淋(S)。Q-SUN Xe-3-HDSE 马上询价
Q-SUN Xe-3-HSCE氙灯耐候试验箱同时具有湿度控制、喷淋以及制冷功能，其命名规则为湿度(H)、喷淋(S)、制冷(C)。Q-SUN Xe-3-HSCE 马上询价
Q-SUN Xe-3-HE氙灯加速老化箱是一款经济的氙灯加速老化箱，具备有湿度控制功能，但是该型号的产品不具备水喷淋功能。Q-SUN Xe-3-HE 马上询价